Country of Origin︰China
Unit Price︰US $ 1 / pc
Minimum Order︰1 pc
1. Power Supply Condition
Power Supply Voltage: AC220V±22V
Power Supply Frequency: 50Hz±0.5Hz
Power Supply Capacity: ≥5kVA
Power Supply Allowed Max. Internal Resistance: 1Ω
2. Radiography Condition
Tube Voltage: 40~90kV, total 11 steps, 5kV per step.
Tube Current: 30, 50mA, total 2 steps.
Time: 0.05s~6s, total 20steps.
3. Mechanical Property
The distance moving range from focus to ground surface when the x-ra
y tube window is down: 450mm~1770mm
X-Ray Tube Circle Horizontal Axis Angle: ±180°
X-Ray Tube Circle Vertical Axis Rotating Angle: ±90°
Bedside mobile design;
Wireless remote control
Payment Terms︰ TT/LC/DP/DA